- Nonin Medical Port Devices Driver Download For Windows 7
- Nonin Medical Port Devices Driver Download For Windows 8.1
- Nonin Medical Port Devices Driver Download For Windows 10
- Nonin Medical Port Devices Driver Download For Windows Xp
Download Box apps on all your devices: Mac, Windows, iPhone, Android, for seamless collaboration and security that satisfies even the most regulated industries. To download data, you need a compatible Nonin computer cable. Connect the appropriate end of the Nonin computer cable to the pulse oximeter and the other end to the PC’s communications port. If using an extension cable, it is necessary to disconnect this prior to downloading.
Norav Medical - ECG Devices and Solutions for All Your Cardiology Needs. ECG Recorders, Blood Pressure Monitors, and Information System Software. The Latest Technology - Wireless & Wired Options. Compatible with Smartphone, Tablet or Desktop. 25 Years in Cardiology. Lifetime-Free Software Updates.
Legacy download files
Legacy PROWIN32 and PROWIN64 download packages provide PROSet, driver, ANS, FCoE, and DCB support for Windows 7* and Windows Server 2008 R2*.
Regular webpacks provide Intel® PROSet support for operating systems Windows 8.1*, Windows Server 2012*, Windows Server 2012 R2*, Windows® 10, Windows Server 2016*, or newer supported operating systems.
Note: New hardware support or new features will not be added to legacy download packages.
This download contains the Intel® Ethernet network drivers and software for Windows 7*.
Which file should you download?
Note: 10GbE adapters are only supported by 64-bit drivers:
- PROWin32.exe for 32-bit (x86) editions of Windows*
- PROWinx64.exe for 64-bit (x64) editions of Windows
How to use this download
Download the self-extracting archive and run it. It will extract the files to a temporary directory, run the installation wizard, and remove the temporary files when the installation is complete. All language files are embedded in this archive. You do not need to download an extra language pack.
See readme notes if you want to extract the files without installing.
This software may also apply to Intel® Ethernet Controllers. Support for built-in network connections is provided by the system or board manufacturer.
About Intel® drivers
The driver or software for your Intel® component might have been changed or replaced by the computer manufacturer. We recommend you work with your computer manufacturer before installing our driver so you don’t lose features or customizations.
See list of manufacturers’ Support websites.
Operator's Manual - WristOx2 3150 with USB View PDF
Utilities and Drivers - WristOx2 3150 with USB View Folder
Operating Instructions - 3150 Memory Clear Application View PDF
WristOx2 3150 Pulse Oximeter: Use/Battery Change View Video
Spec Sheet - WristOx2 Model 3150 USB View PDF
Disposable Wrist Strap Instructions - WristOx2 Model 3150 USB View PDF
WristOx2 for Sleep Studies View PDF
WristOx2 for Ambulatory Monitoring View PDF
Sensor Compatibility Guide View PDF
Understanding Accuracy, Bias, and Variability in SpO2 Readings View
PureSAT SpO2 Averaging View
Nonin Medical’s WristOx2® Model 3150 Beats VirtuOx VPOD in Hypoxia Testing View
Accuracy and Superior Performance of PureSAT and PureLight Oximetry Technologies View
Using Overnight Pulse Oximetry to Manage Oral Appliance Therapy (OAT) during Treatment for Obstructive Sleep Apnea View
Accurate Oximetry Monitoring – See the Technology Difference View
Pulse Oximetry at Home View
WristOx2® Model 3150 with USB Specifications
Oxygen Saturation Display Range: 0% to 100% SpO2
Pulse Rate Display Range: 18 to 321 beats per minute (BPM)
Numeric: 3-digit LCD
Pulse Strength: Pulse Strength Bar Graph
Accuracy – Sensors: Declared accuracy data for compatible sensors can be found in Nonin’s Sensor Accuracy document.
Measurement Wavelengths and Output Power**:
Red: 660 nanometers @ 0.8 mW maximum average
Infrared: 910 nanometers @ 1.2 mW maximum average
** This information is especially useful for clinicians performing photodynamic therapy.
Operating: -5 °C to 40 °C (23 °F to 104 °F)
Storage/Transportation: -40 °C to 70 °C (40 °F to 158 °F)
Time (from storage) for monitor to be ready for its intended use: 10 minutes to warm from -40 °C to -5 °C
10 minutes to cool from 70 °C to 40 °C
Device temperature will not exceed 41°C as measured during a controlled environment test.
Operating: 10% to 95% non-condensing
Storage/Transportation: 10% to 95% non-condensing
Operating Altitude: Up to 4,000 meters (13,123 feet)
Operating Hyperbaric Pressure: Up to 4 atmospheres
Power Requirements: Two AAA (1.5V) batteries
|Battery Life (expected minimum):|
NOTE: Based on testing new and fully-charged batteries. See
footnotes for brands used. Refer to battery manufacturers’
operator’s manuals for instructions for use.
|Alkaline AAAa)||Rechargeable AAA (700 mAh)b||Rechargeable AAA (1050 mAh)c|
|Storage: MVI Display Mode off:|
MVI Display Mode on:
|Not specified||Not specified|
|Operating without Bluetooth Connection, continuous use:||53 hours||36 hours||52 hours|
|Operating with Bluetooth Connection, continuous use:||44 hours||24 hours||31 hours|
Dimensions (without sensor or wristband): 51 mm x 73 mm x 19 mm (H x W x D)
(2.0 in. x 2.9 in. x 0.75 in.)
Weight (with batteries and wristband): 70.0 g (2.5 oz)
up to 1,080 hours (4 sec. data storage rate)
up to 540 hours (2 sec. data storage rate)
up to 270 hours (1 sec. data storage rate)
Classification per ANSI/AAMI ES60601-1 and CAN/CSA-C22.2 No. 60601-1:
Type of Protection: Internally powered (battery power)
Degree of Protection: Type BF-Applied Part
Mode of Operation: Continuous
Enclosure Degree of Ingress Protection: IP33
This product complies with ISO 10993-1, Biological evaluation of medical devices – Part 1: Evaluation
a.Batteries used: Harding Alkaline AAA
b.Batteries used: Energizer (MR03) Ni-MH 700mAh Rechargable Batteries 1.2 VDC
c.Batteries used: Ansmann (HR03) Ni-MH 1050 nMh Rechargable Batteries 1.2 VDC
WristOx2® Model 3150 with USB Troubleshooting
|Problem||Possible Cause||Possible Solution|
|Device will not activate.||Batteries inserted wrong.||Check batteries.|
|Batteries are depleted.||Replace batteries.|
|Sensor is disconnected.||Reconnect sensor.|
|Device is in Sensor Activation mode and has timed out.||Press the activation switch.|
|Disconnect and then reconnect the sensor.|
|Device is in Programmed mode.||Use nVISION software to select Spot Check or Sensor Activation mode.|
|%SpO2 and pulse rate do not display.||Device set in Partial Display mode.||Use nVISION software to select Full Display mode. Reconnect sensor.|
|Poor pulse signal indicator displays.||Excessive patient motion.||Reduce patient motion.|
|Poor pulse signal indicator displays and pulse strength indicator shows two bars or less.||Inadequate pulse signal.||Reposition or replace sensor, or place sensor on a different finger.|
|Remove and reconnect sensor.|
|Hands are cold.||Warm sensor application site.|
|No pulse display on pulse strength bar graph indicator.||Sensor applied incorrectly.||Refer to sensor Instructions for Use for proper sensor application.|
|Device needs repair.||Contact Nonin Technical Service.|
|Possible interference from blood flow restrictors (arterial catheters, blood pressure cuffs, infusion lines, etc.).||Reduce or eliminate restriction.|
|Reduced circulation due to excess pressure from sensor.||Check sensor alignment, reposition sensor, verify correct sensor size.|
|Excessive ambient light.||Shield sensor from light source. Check sensor alignment.|
|Problem||Possible Cause||Possible Solution|
|No pulse display on pulse strength indicator (continued).||Sensor applied to polished or artificial nail.||Remove fingernail polish or an artificial nail.|
|Sensor Light-Emitting Diode (LED) is not lit.||Contact Nonin Technical Service.|
|Er 01 displays on LCD.||Device configuration memory failure.||Device reverts to default settings|
(Spot-Check mode, 4-second sample rate). Use nVISION software to change settings. If error code continues, contact Nonin Technical Service.
|Er 02 or 04 displays on LCD.||Device memory failure.||Contact Nonin Technical Service.|
|Er 03 or 05 displays on LCD.||Device failure. Device memory intact, but device may have lost most recent session or stored data.||If error code continues, contact Nonin Technical Service.|
|Dashes continually display on LCD.||Sensor malfunction.||Replace sensor with a Nonin-branded sensor.|
|Device does not record in |
|Data collection start and stop times are set incorrectly.||Use nVISION software to program correct start and stop times.|
|Clock settings are lost after replacing batteries.||Use nVISION software to reset clock.|
|Devices will not pair.||Device is out of range.||Verify device is in range while being paired (approximately 100 meters [328 feet] spherical radius).|
|Bluetooth radio has timed out.||Press activation switch to turn on Bluetooth radio.|
If these solutions do not correct the problem, please contact Nonin Technical Service at (800) 356-8874 (USA and Canada), + 1 (763) 553-9968, or +46 650 401500 (Europe).
Nonin Medical Port Devices Driver Download For Windows 7
For general questions, please contact Nonin Customer Support.
Nonin Medical Port Devices Driver Download For Windows 8.1
For product technical, troubleshooting or repair questions, please contact Nonin Technical Support.
Nonin Medical Port Devices Driver Download For Windows 10
Stay informed with updates from Nonin
Nonin Medical Inc
13700 1st Ave N,
Plymouth, MN 55441
Toll Free (USA and Canada)
Nonin Medical Port Devices Driver Download For Windows Xp
Nonin Medical B.V.
Doctor Paul Janssenweg 150
5026 RH Tilburg